Search found 8 matches
Search found 8 matches • Page 1 of 1
- Wed May 28, 2014 6:05 pm
- Forum: BioSemi all other topics
- Topic: Locator/Patriot in Windows 7
- Replies: 0
- Views: 11026
Hi All, I was wondering if any of you have experience using the EMSE program called Locator in Windows 7 with the Patriot hardware. I ask on this forum because I've talked to customer support at EMSE and they didn't seem to think that many (if any) people have been using this program on Windows 7. C...
- Wed Feb 26, 2014 3:32 pm
- Forum: BioSemi Software (ActiView executable)
- Topic: Computer Crashes and Actiview
- Replies: 0
- Views: 10167
Hello, I've been asked to inquire about possible tips on how to avoid computer crashes while recording EEG sessions in ActiView. Just last week we had our EEG acquisition computer crash in the middle of a task and we were wondering if there are any secrets from the pros on how to avoid having this h...
It looks like I was mistaken. Replacing the fuse did change the situation, now the green light does activate when the battery is turned on but the light on the collection box does not illuminate to indicate that it is receiving power when it is connected to the battery. The collection box works just...
Hello, We've recently been having issues in our lab with a battery that does not appear to be working correctly. Whenever the battery is turned on the green light indicating that it is powered on does not activate, nor does the collection box indicate that it is receiving power from the battery when...
- Mon Jul 23, 2012 6:09 pm
- Forum: BioSemi Hardware
- Topic: The Use of Tap Water When Cleaning Electrodes
- Replies: 2
- Views: 3542
Hello BioSemi. I've noticed in other posts that tap water has been approved for cleaning electrodes, but what about if the water is visibly contaminated with iron/rust? I'm in the process of setting up a new EEG lab and the water that we would use to clean the electrodes has an orange-ish tint. Cons...